- SleepCube device
- Carrying case
- Spare filter pack
- Air supply tubing
- AC line cord
- User manual
Frequently Asked Questions:
Do I need a voltage converter to use my SleepCube in a foreign country?
A voltage converter is not needed because the SleepCube range of CPAP’s have a switch-mode power supply, which automatically adjusts to voltages from 100—240 volts and 50—60 Hz. It is, however, necessary to purchase the correct mains lead for the country you are travelling to.
Can I use my SleepCube in a motor home, vehicle, boat?
The SleepCube series CPAP’s can all be operated via a 12V DC power source. There are two 12 V DC operating options available: a 12 V DC accessory cable for a cigarette lighter in a car (DV51D-619) and a 12 V DC accessory cable for a stand alone battery (DV51D-696). Please Note: If heated humidification is used, the heater will NOT operate directly from a 12 V DC source. DC power can be passed through an inverter that will deliver AC power to the system (required 400 Watt / 230 V).
As a patient, can I make changes to my CPAP pressure?
The SleepCube range of CPAP’s are only available with a prescription from a sleep physician. The pressure that your SleepCube has been set to will have been tailored to your requirements as determined by sleep screening tests. Any pressure changes should be made by your physician upon reassessment.
I intend on travelling, will I require my SleepCube to be re-adjusted for large changes in altitude?
No. The SleepCube series CPAP’s contain a pressure transducer which allows the unit to automatically adjust for altitude changes up to 2,700 m (9000ft).
My doctor has advised that I need to have supplemental oxygen at night whilst receiving my CPAP therapy, is this possible?
Yes, supplemental oxygen can be added with the optional oxygen adapter attached between the unit and tubing (7353D-601) or if the mask is equipped with an oxygen port directly to the mask.
What mask should I use with my SleepCube CPAP?
The SleepCube can be used with any vented mask. The success of CPAP therapy is often dependant upon the comfort and fit of the mask so it is important to have a wide choice. Most suppliers of sleep therapy equipment have a range of masks available including nasal masks, full face interfaces, nasal pillows and mouth only masks.
What length of tubing can I use with my SleepCube?
The SleepCube can be used with either the standard 6ft tubing or 10ft tubing (DV51D-630).
I experience a sore, dry throat and nose. Is there anything I can do?
The most common solution is a heated humidifier. This is a device that fits to the CPAP to heat and moisten the airflow making therapy more comfortable. The SleepCube range of CPAP’s has an integrated heated humidifier available as an accessory. For more information and advice please refer to your sleep specialist.